Cofoe Test Cholesterol meter & Uric acid meter & blood glucose meter 3 in 1 multi monitoring meter, test strips & Lancets
✔ One machine, three uses：blood glucose, uric acid, total cholesterol
✔ Quick output value：blood glucose, uric acid output value in 6-150s
✔ Large LCD HD screen
✔ Micro blood collection
✔ 300 sets of memory
✔ 5 years warranty
The monitoring system of blood sugar, uric acid and total cholesterol
is designed to test the concentration of total blood cholesterol,
blood sugar or uric acid in the whole blood. It is not suitable for the detection of newborn. The test paper can only be used to detect the concentration of total cholesterol, blood sugar, or uric acid in the fresh blood of the fingertips.
Note：we had english manual if you need !
1) Glucose test:
If the fasting blood glucose and the blood glucose within 2 hours after meal are higher than normal, diabetes is suspected
2) Total cholesterol:
High total cholesterol represents high blood fat, which causes insulin resistance. Insulin resistance itself increases cholesterol, and once insulin levels drop, blood sugar increases.
3) Uric acid:
It is for gout reference, if the uric acid is too high in the human body then gout occurs.Therefore, this product can accurately measure the uric acid value to determine whether gout exists.
① Meter and 3 in 1 test strips=1 Blood glucose meter +1 Lancing pen +25 pcs glucose strips +10 pcs cholesterol strips+25pcs uric acid strips+60pcs needles
② 25 glucose strips=25 pcs glucose strips +25pcs needles
③ 25 Uric Acid strips=25pcs uric acid strips+25pcs needles
④ 10 cholesterol strip=10 pcs cholesterol strips+10pcs needles
⑤Only a monitor=1 monitor +1 lancing pen
1. Unscrew the front cover of the blood collection pen counterclockwise, place the base of the blood collection needle on the needle holder, and make sure that it is properly installed.
2. Screw on the front cover of the blood collection pen and adjust the blood sampling depth.
3. Remove a test strip from the test strip and immediately cover the test strip to keep other test strips dry.
4. Insert the test paper contact line part into the analyzer's detection port, the analyzer will automatically start.
5. Make sure that the code on the screen is the same as that on the test strip. When a drop of water is displayed on the screen, the analyzer is ready for inspection.
Attention: If drop didn’t appear, please remove unused test strips and reinsert them.
6. Pull the blood collection lever backwards to prepare for blood
7. Put the blood collection pen against the blood collection site
Press the back button
8. Wipe off the first drop of blood, do not use.
Gently massage your fingers to form a round blood drop.
Attention: Make sure that you have enough blood. If the blood is spread out, please wipe it and press the new round blood drop again.
9. Touch the specimen into the end and touch the blood.
Blood will automatically enter the blood.
Attention : Hearing hissing after entering the blood indicates successful blood flow.
Attention: Unsuccessful blood infusion can lead to inaccurate or failed test results.
6-150 s After the countdown, the result is displayed on the screen.
Discard used test paper in biomedical waste containers.
Analyzer will automatically shut down